Certified Pre - Owned
The AirCurve™ 10 VAuto BiLevel Machine with HumidAir™ Heated Humidifier uses advanced technology to aim for therapy comfort on many levels. This is an auto bilevel machine with a built in heated humidifier capable of self-adjusting the humidification level based on the temperature of the room.
The machine automatically adjusts inhalation and exhalation pressure on a breath by breath basis while monitoring the length of inhalation breaths and responding to regulate the breathing cycle.
1 Year Warranty
- AutoSet Algorithm
- Built In HumidAir™ Humidifier
- Optional ClimateLineAir™ Heated Tube
- Expiratory Pressure Relief (EPR™)
- Vsync Leak Management
- TiControl Feature
- My Options Color LCD Display
- Light Sensor
- Advanced Data
ResMed developed an advanced algorithm technology for bilevel therapy to continuously monitor breathing in order to adjust the inhalation and exhalation pressure breath by breath. Adjusting the delivered pressure based on the needs of the user increases the comfort of the therapy while maintaining an open airway.
HumidAir™ Heated Humidifier and Optional ClimateLineAir™ Heated Tube
The HumidAir™ humidifier is built into the AirCurve™ 10 machine taking up limited space on a bedside table. Easily adjust the humidification level using the My Options screen on the machine. To set the amount of moisture added to the air stream, select levely 1 through 8 with 1 being the lowest level of humidification, and 8 being the highest. If humidification is not desired, turn the humidifier off entirely.
The OPTIONAL ClimateLineAir™ Heated Tube can be added to the therapy system to optimize humidification. Select the "Auto" function on the AirCurve 10 VAuto to allow the Climate Control system to set for the optimum humidification with minimal rainout condensation. The system considers the temperature of the room air to deliver the optimum level of humidification. Please Note: The ClimateLineAir™ tube is 78 inches in length.
Expiratory Pressure Relief (EPR™)
When the AirCurve™ 10 VAuto is set in S Mode, the machine is designed to offer expiratory pressure relief to help therapy feel more natural. ResMed's EPR™ feature adjusts the amount of pressure delivered during exhalation and inhalation. EPR™ can be turned off, or set to 1, 2, or 3. Level 1 is the smallest change between inhalation and exhalation and level 3 is the largest change in pressures.
Vsync monitors mask leaks throughout the night and compensates by adjusting the baseline pressure to maintain therapy effectiveness.
The TiControl feature on the AirCurve VAuto machine monitors the pattern of inhalation and exhalation, comparing it against the TiMax and TiMin settings on the machine. Those with respiratory disorders such as COPD or obstructive lung disease may experience early or delayed changes from inspiration to expiration, referred to as cycling. The 10 VAuto BiLevel TiControl feature is designed to prevent cycling by setting a minimum and maximum time limit on inspiration. When the machine detects the user breathing which falls outside the machine settings, it will intervene to alter the timing of the delivered IPAP (inhalation) or EPAP (exhalation) pressure.
- Ti Max: Maximum length of Inhalation
- Ti Min: Minimum length of Inhalation
- Trigger: Sensitivity level for inhalation trigger
- Cycle: Sensitivity level for exhalation
IMPORTANT FUNCTION NOTE: The AirCurve 10 VAuto machine does spontaneously regulate the breathing pattern to avoid cycling which results in inefficient breathing. This feature CANNOT be turned off. This machine is NOT a BiLevel ST machine and DOES NOT have a back up rate setting.
LCD Display with My Options
The AirCurve™ 10 VAuto has a color LCD display. The menus are designed to be easy to read and use. Select the menu item on the screen by turning the easy grip dial. Select the My Options menu to view and modify the current settings:
- Ramp time - Ease into therapy with a period of lower pressure. Or select AutoRamp™ and the machine senses and adjusts when sleep begins.
- Humidity Level - Set humidity between 1 (lowest) & 8 (highest), or turn it off.
- Mask Fit - Runs a seal integrity check showing a green smiling face for a good fit or red frowning face for a poor fit.
- Leak Alert- Set the device to begin beeping when a large leak is detected or the mask is removed.
- SmartStart™ - Therapy automatically starts when it senses breathing.
- Mask Type - Set the mask type being used: Nasal; Nasal Pillow; Full Face Mask.
- Run Warmup - Preheat the humidifier water before starting therapy.
The AirCurve™ 10 includes an integrated light sensor to adjust the machine screen light level. The sensor reads the light in the room and adjusts the screen brightness automatically. Once the lights in the room are turned off, the machine screen dims. After a few seconds the machine screen goes dark.
The AirCurve™ VAuto records the therapy information on the machine's on board memory. The data can be downloaded to the included SD card. The machine's on board memory holds compliance information for 365 sessions of use as well as detailed data for 30 sessions.
Machine Operating Modes: CPAP, BiLevel, Auto-BiLevel
Machine Pressure Range: 4 to 25 cm H2O
Machine Pressure Increment: .2 cm H2O
Machine Dimensions: 4.57 x 10.04 x 5.91 Inches with Chamber; 4.57 x 8.00 x 5.91 with Optional Side Cover (Approximate)
Machine Weight: 2.75 Pounds with Chamber; 2.45 Pounds with Optional Side Cover (Approximate)
Tubing Compatibility: All 22mm Standard Tubing, All 15mm Slim Standard Tubing, ResMed SlimLine Tubing, and ResMed ClimateLineAir Heated Tubing
Operating Altitude: Sea Level to 8500 Feet
Operating Sound Levels: 26.6 dBA (Average)
Filters: Disposable Standard and (Optional) HypoAllergenic
Comfort Technologies: Easy-Breathe, EPR (Depending on Current Device Mode), Climate Control (when ClimateLineAir Heated Tubing is Present)
Leak Compensation: Automatic
Pressure Ramp: Standard Ramp User Adjustable 0 to 45 Minutes
Pressure Ramp Start: 4 to Minimum Therapy Pressure Setting
Altitude Adjustment: Automatic
Automatic Start & Stop: Yes, SmartStart & Stop
Supplemental Oxygen Use: Compatible up to 4 LPM with Adapter (Compatible up to 15LPM in CPAP or BiLevel S Mode with Adapter)
Display: Color LCD Display with Ambient Light Sensor
Humidifier: Built-In HumidAir Humidifier
Humidifier Settings: 1 (Lowest) to 8 (Highest) and Off
Humidifier Chamber Capacity: 380 mL to Max Fill Line
Humidifier Chamber Type(s): Standard and (Optional) Long Life Dishwasher Safe
Battery Power: (Optional) External ResMed Power Station II, Long Life or Lightweight Travel Battery Pack
Data Storage: SD Data Card
Data Viewing: On Screen Sleep Report, MyAir Online Interface, ResScan Software (Not Available to Consumers)
Data Transmission Options: SD Data Card, Built-In Modem
FAA Compliant Label Included on Base of Unit. This product is approved by the FAA for use on all commercial flights within the US and all international flights that depart or arrive in the US. Always contact your airline prior to your flight to confirm any requirements they may have -- such as a copy of your prescription or a letter of medical necessity -- for allowing your device on board.
- AirCurve 10 V-Auto with Built-In HumidAir Heated Humidifier
- Standard Humidifier Chamber
- Breathing Tube
- Universal AC Power Supply with Cord
- Travel Bag
- Filter Cover
- Printed User Manual
Refurbished Units Can not be linked to My Air App or AirView
SD Card NOT Included
Recommended Replacement Schedule
|Full Face Mask||1 every 3 months||The mask is the plastic portion of the interface. That part should be replaced every 3 months. The cushion part is different. The cushion inserts have to be replaced more often because of the contact with the face. They can be separated from the frame work and should be replaced once a month.|
|Nasal or Pillow Mask||1 every 3 months||The mask is the plastic portion of the interface. That part should be replaced every 3 months. The cushion part is different. The cushion inserts or pillows have to be replaced more often because they touch the face. They can be separated from the frame work and should be replaced once a month.|
|Headgear||1 every 6 months||CPAP mask headgear and chin straps may become stretched and lose their elasticity, leading to over-tightening and discomfort. They can also trap bacteria from sweat and moisture.|
|Tubing||1 every 3 months||CPAP mask tubing may develop small holes or tears, which can cause air leaks. If your tube is leaking, you may not be receiving your prescribed therapy setting from your CPAP. This can cause you to feel like you’re not sleeping as well.|
|Chinstrap||1 every 6 months||Chinstraps may become stretched and lose their elasticity, leading to over-tightening and discomfort. They can also trap bacteria from sweat and moisture.|
|Full Face Mask Cushion||1 every month||Full face mask cushions cover more of your face and may deteriorate over time through regular wear and tear and cause leaks throughout treatment. Regular replacement is recommended for hygienic purposes and to ensure the best fit and most effective treatment.|
|Nasal Mask Cushion||2 every month||Nasal mask cushions are in constant contact with your face and should be replaced consistently. Regular replacement is recommended for hygienic purposes and to ensure the best fit and most effective treatment.|
|Pillow Mask Pillows||2 every month||Pillow masks are in constant contact with your face and the pillows sit inside your nose throughout the night. Regular replacement is recommended for hygienic purposes and to ensure the best fit and most effective treatment.|
|Heated Tubing||1 every 3 months||CPAP mask tubing may develop small holes or tears, which can cause air leaks. If your tube is leaking, you may not be receiving your prescribed therapy setting from your CPAP. This can cause you to feel like you’re not sleeping as well.|
|Disposable Filters||2 every month||CPAP device filters can wear out or become clogged over time, potentially exposing you to airborne particles, mold and bacteria.
|Non-disposable Filters||1 every 6 months||CPAP device filters can wear out or become clogged over time, potentially exposing you to airborne particles, mold and bacteria.|
|Humidifier Water Chamber||1 every 6 months||CPAP humidifier water chambers may become discolored, cracked, cloudy or even pitted due to the mineral levels found in most tap and drinking water. As the material deteriorates, cracks may trap bacteria from moisture.|
|Mask Frame System||1 every 3 months||The frame system is comprised of the mask frame and cushion. Due to being in close proximity to the face during treatment, frames should be regularly replaced. However, since not in direct contact, the frame system is only eligible for replacement every three months.|
|CPAP/BiPAP Machine||1 every 5 years||A CPAP machine's life span is typically 5 years. We recommend getting a backup CPAP machine, especially if your current machine is more than two years old. That way you’ll never be without your therapy, even if one of your devices requires maintenance.|
Payment & Security
Your payment information is processed securely. We do not store credit card details nor have access to your credit card information.